3D Optical Profilers
Position Metrology Systems
Custom Metrology Solutions
Trade-in & Upgrade Program
Applications Form & Roughness of Precision Parts Form, Waviness, Roughness Surface Finish Regions Analysis New! Tribology Automated Image Inspection Step Height Thin Film Measurement Transmitted Wavefront Radius of Curvature Optical Homogeneity Position & Angle Measurement
Measuring Surface Finish
Measuring Surface Finish - Crosshatch Machining Marks
Measuring Surface Finish - Prosthetic Knee Joint Surface
Stylus Profilers for Measuring Surface FinishStylus profilers are contact profilers in that they physically touch the surface being measured, typically with a diamond tip on a delicately balanced arm. The stylus is dragged in contact with the surface, and as the diamond tip encounters peaks and valleys, the tip is raised and lowered from an otherwise straight line. The amount to which the stylus is raised or lowered at any given point is recorded. Sometimes the variation is compared to a reference surface, or sometimes the motion is detected in relation to some reference point in comparison to the stylus arm's position.
The result from a stylus measurement is often expressed as a single parameter (Ra), but this number is calculated from a linear data set, which is normally graphed as a line - in a profile plot.
Measuring Surface Finish with Non-contact ProfilersNon-contact methods of measuring roughness or surface finish are also common, such as Coherence Scanning Interferometry (CSI), which is used by ZYGO's lines of Optical Profilers; confocal microscopy techniques; focus variation; structured light techniques such as ZYGO's Moire interferometry; or using capacitors as a proximity sensor, measuring induced fluctuations in an electrical field.
Reasons non-contact surface finish measurements may be preferred or required:
• If a material is soft, fragile, pristine, or chemically sensitive and won't tolerate contact methods
• If material lay is a factor, and a roughness measurement in more than one direction is desired
• If consumables, such as fragile diamond stylus tips, are difficult to maintain in inventory - they aren't needed for an optical profilometer
• If volumetric data is required, such as when measuring wear of materials or the volume of an engineered micro structure
Inquiry FormPlease use the form below to contact us with any questions you have regarding Measuring Surface Finish, or any other ZYGO metrology applications.
** You may withdraw this consent at any time using the "unsubscribe" link at the bottom of any page on this web site.
Contact ZYGO Today
× Enter the email address you would like to unsubscribe from future contact...